This comment was originally posted in this thread by Joniale
It seems there is an error in the ECAT report. Could somebody confirm this?
Please, see the comments of Mats in the following article
Here a detailed explanation of the problem. If somebody with the knowledge can explain.
They are discussing about an inverted clamp that can cause the COP 3.
Please i need to know if this is destroying the report. 🙁